Diamond Burs For Drilling Teeth Treatment Green (C) Coarse / 125-150μ
Detailed Product Description:
||CE / ISO / FDA / FSC
||Diamond + Stainless Steel
||5 Pcs Per Blister
||Diamond Burs Show Book
||Used connecting with dental handpiece
Diamond Burs Used In Dentistry
1. The usage of natural diamond powder, which makes diamond has a very high hardness and excellent physical and chemical features .
2. It made as diamond burs finally with certain cutting tissue adhesion or plating them with a certain angle of diamond particles, according to a certain direction, distance to the center of the stainless steel burs.
3. The product is used for the treatment of drilling teeth in the oral cavity .
4. The burs is made of a pin handle and a pin.
Specifications and models:
Flat head conical , apex type flame , taper head , pointed cone , cone round nose cone type , flat type , column type , round ball , ball spire tight mouth , single inverted cone, rugby, pear type, round type.
Classification coding system reflects shapes of burs.
TF / Taper Flat End TR / Taper Round End TC / Taper Conical End
FO / Flame Ogival End SF / Straight Flat End SO / Straight Ogival End
SR / Straight Round End RS / Rounded Shoulder EX / Special (Extra) Shape
BR / Ball Round Type BC / Ball Collar Type DI / Double Inverted Cone
SI / Single Inverted Cone WR / Wheel Round Edge CR / CR Inlay Preparation Dia-Burs
CD / Children's Dentistry Dia-Burs S / Short Shank SS / Super Short Shank
C / Coarse F / Fine EF / Extra Fine
Main performance and technical indicators:
1. Shank hardness not less than 250hv0.5 .
2. The surface roughness R of the shank of the car should be no greater than 0.4mu m .
3. Corrosion resistance and corrosion resistance test shall not produce corrosion and deformation of the function, but it is allowed to have a slight electrical corrosion in the head and neck .
4. The radial circle of the needle should be no greater than 0.08 mm .
Diamond Burs For Crown Preparation
Use of instructions and precautions:
1. Sterilization should be sterilized with high temperature steam before using the needle .
2. Be careful to place the pin on the head of the mobile phone according to the operation instructions of the mobile phone .
3. When the operation is used , when the needle is rotated , it will be pressed to the part where it is to be drilled or polished .The pressure of the needle is 50 ~ 100g is appropriate , the pressure should not be too large , and requires the force to be uniform .
Storage and duration:
After packing , the needle should be stored in a room with relative humidity less than 80% , no corrosive gas and good ventilation . Under the condition that the needle is packed and abide by the storage rules , it shall be guaranteed not to rust in a year and a half .
The work head shall be specified in the parts according to the following figures .
D1: the diameter of the working part , the maximum diameter of the head or part of the working part .
D2: diameter of the neck , indicating the diameter of the shank of the emery .
L1 : working length , refers to the coating emery working length of the part .
L2 : refers to the entire bur the length of the needle.
|Dental Root Canal Treatment
||Hospital , Clinic , lab...
1. Diamond bur has a stainless steel working part which is coated with fine diamond crystal .
2. It is defined as the rotary grinding instruments for hard tissues such as tooth and bones .
3. It is used connecting with dental handpiece .
4. It can also be used for grinding metal, plastic , porcelain and other hard materials.
Diamond Burs Color Coding